* Auto-attenuated alternating magnetic field
* Low-frequency alternating magnetic field
<Control Panel Introduction>
The panel is divided into three main blocks
* Tesla Meter Control* Demagnetizer Control* Power Switch
<Instructions of Stationary Demagnetizing Machine>
a. Place the workpiece in the designated area
b. Adjusting the demagnetization parameters:
Turn the left knob to adjust the demagnetization magnetic field (turn right to increase).
Turn the right knob to adjust the demagnetization time (turn right to increase).
c. Start & Stop
<Instructions of Tesla Meter>
Note: The Tesla meter can only be operated during standby mode.
a. Insert the Tesla sensor probe, and move the probe to the point to be measured on the workpiece.
b. Switch on the "Real / Hold" button to hold the highest value among the measured values.
c. Use "G / mT" button to select G or mT as the measurement unit
d. Use the "Zero" button to subtract the ambient magnetic field.
Placing the workpiece in the designated area to demagnetize.
<Combination of Multiple Machines>
Can be combined to demagnetize larger workpieces.
Examples of demagnetizing with 3 HD-SL1000s.
豪昱電子有限公司HOLDING ELECTRIC CO., LTD.